The Tramex IQCK is a compact kit for instant, non-destructive, quantitative moisture content testing of concrete floors and ambient dew-point conditions at the time of installation. Even if pre-installation tests have been performed, checking the %MC and ambient dew-point conditions at the time of installation is crucial to ensure successful installation. The Tramex moisture meter allows for non-destructive quantitative testing of concrete moisture content using Gravimetric testing as a baseline. The provides surface temperature readings and ambient temperature and dew point conditions to avoid condensation-related issues at the time of installation and floor failures. The allows for data reporting to mobile devices, to GeoTag test results, create, save and share photos, reports and charts. The Tramex Flooring Installation Quick Check Kit comes in a Tramex Easy-travel case.
Concrete Moisture Encounter 5 - CME5 - Measuring Range:
|Depth of penetration:||approx. ¾" (20 mm)|
|Moisture Content for Concrete:||0 to 6%|
|Comparative for Gypsum floor screed:||0 to 10|
|Reference scale:||0 to 100|
Infrared Thermometer and Dew Point Detector - IRT2DP - Measuring Range:
|Temperature Range:||-58ºF to 716ºF (50°C to 380°C)|
|Emissivity:||Fixed at 0.95|
|Temperature accuracy:||-58º to 32ºF / -50º to 20ºC: +/- 32º to 716ºF / 3.5ºC 20º to 380 ºC: +/-1.5ºC|
|Wavelength range:||8 – 14um|
|Humidity Range:||0 - 100% RH|
|Humidity accuracy:||+/-3.5% (20% to 80%)|
|Dew point temperature:||(-22 to 212)ºF / (-30 to 100)ºC|
|Operating temp:||32 to 122ºF (0 to 50ºC)|
|Storage temp:||-4 to 140ºF (-10 to 60ºC)|
|Relative humidity:||10%-90%RH operating, <80%RH storage|
|Power Supply:||9V battery, NEDA 1604A or IEC 6LR61, or equivalent|
Specifications of Kit:
|Size:||13.2” x 9.5” x 3.2” (33.5cm x 24cm x 8cm)|
|Power:||2 x AA batteries, 1 x 9 volt PP3 battery|
Calibration Check Plate for Tramex , , CME4 and CMEX2. The calibration check plate allows for regular on-site calibration checks.
The Extension Handle can be used with the , , and Moisture Meters. It is a useful accessory for measuring in out of reach, high places. It can also be used for taking readings on the floor and thus reduces the need for crouching or kneeling.