The Tramex Hygrohood is an insulated humidity hood with built-in hygrometer used for non-destructive Relative Humidity testing of concrete and other floors and screeds to international standards.
Manufacturers or national standard recommendations:
British Standards code of practice BS8201, BS8203,BS5325 suggests that a concrete floor or screed shouldbe sufficiently dry to allow installation of a resilient floorcovering and tested using the insulated impermeablehood method.Non-destructive moisture content tests with the TramexCME4 or CMEX2 can be carried out until the floor orscreed reaches the moisture level specified by thefloor-covering manufacturer. The Tramex CME4 orCMEX2 can be used to determine the areas of greatestconcern on the floor or screed. At that stage relativehumidity tests using the Hygrohood can be implementedto corroborate the non-destructive moisture contenttest.
Pre-test conditioning and preparation:
For best and most accurate results, tests should be carried out after the internal conditions of the building in which the slab is located have been at normal service temperature and humidity for at least 48 hours. All artificial heating or drying equipment should be turned off at least 96 hours before final readings are attempted, otherwise results may not accurately reflect the amount of moisture present or moisture movement in the slab during normal operating conditions. Avoid testing in locations subject to direct sunlight or sources of heat. Use of artificial aids for accelerated drying of concrete is not recommended. If they are being used it is recommended they should be turned off at least four days before taking final readings. It is advantageous to know the background of the site e.g. when the floor or screed was poured, thickness levels, etc.
1. Before positioning the Hygrohood on the floor slab, the surface should be clear of any foreign materials and swept clean of any dust or loose materials that could affect a proper seal between the hood and the surface of the floor.
2. Using butyl tape, seal the insulated Hood to the concrete surface.
3. The Hygrometer will record minimum and maximum values while it remains switched on. Optional C°/F° units are available. Memory will be reset when the unit is switched off.
Always refer to the adhesive and/or floor covering manufacturer’s recommendations for the acceptable moisture content levels of concrete floor screeds....More info
The Hygrometer is factory calibrated to 75% RH at 20°C. The calibration should remain stable for a minimum period one year from date of purchase unless the unit is physically damaged or misused.The measuringrange is 20% - 99%RH, with a temperature range of 32° - 122°F / 0°C - 50°C. Tolerances: +/- 3%, +/- 2°F / +/- 1°C. Replacement battery: CR2032.